Home / Store / Journals / Electronic Articles / Nuclear Science and Engineering / Volume 72 / Number 1
S. M. Wu, M. C. Hsu, M. C. Chow
Nuclear Science and Engineering
Volume 72 / Number 1 / October 1979 / Pages 84-96
Format:electronic copy (download)
A new modeling technique is introduced for on-line sensor time constant identification, both for the resistance temperature detector (RTD) and for the pressure sensor using power plant operational data. The sensor's time constant is estimated from a real characteristic root of the fitted autoregressive moving average model The RTD's time constant values were identified to be 8.4 s, with a standard deviation of 1.2 s. The pressure sensor time constant was identified to be 28.6 ms, with a standard deviation of 3.5 ms.
Your cart is empty.
Home|Invoice Payment|Nuclear Links