Home / Store / Journals / Electronic Articles / Nuclear Technology / Volume 175 / Number 1
Edward J. Waller
Volume 175 / Number 1 / July 2011 / Pages 89-92
Format:electronic copy (download)
Recent nuclear weapons testing in the limit of low-yield detonations has underscored the need to ensure that radiation detection and monitoring equipment can adequately respond to these events. Testing and validating equipment in appropriate reference fields have become difficult since the closing of the NATO primary fission spectra reference at the Aberdeen Proving Ground Fast Burst Reactor facility post-9/11. A simple and low-cost device was designed to perform testing of commercial off-the-shelf neutron detection equipment to the expected spectral shape from a low-yield nuclear weapon. By enclosing an 241AmBe (,n) neutron source within a heavy water-moderated sphere, the general shape of a 1-kiloton standard fission weapon was generated at 1 m, valid between 100 and 2000 keV. The 1-m dose rate expected from this configuration is [approximately]2.16 × 10-10 Svh-1Bq-1 , which is less than one-half of the unshielded dose rate.
Your cart is empty.
Home|Invoice Payment|Nuclear Links