Home / Store / Journals / Electronic Articles / Fusion Science and Technology / Volume 58 / Number 1 / Pages 345-351
I. Yamada, K. Narihara, H. Funaba, T. Minami, H. Hayashi, T. Kohmoto, LHD Experiment Group
Fusion Science and Technology / Volume 58 / Number 1 / Pages 345-351
Format:electronic copy (download)
The Large Helical Device (LHD) Thomson scattering system measures electron temperature and density profiles of LHD plasmas along the LHD major radius at a horizontally elongated section. The LHD Thomson scattering system has an oblique backward-scattering configuration. The number of observation points and typical spatial resolution are 144 and 17 mm, respectively. The temporal sampling frequency is 10 to 100 Hz. Measurable temperature and density ranges are Te = 5 eV to 20 keV and ne 1018 m-3 , respectively. The LHD Thomson scattering system consists of several subsystems: laser system, light collection optics, polychromators, and data acquisition system. In the past decade, we have continued our efforts to improve the performance and reliability of the LHD Thomson scattering system through extension of measurable temperature and density ranges, Raman and Rayleigh calibrations for absolute density measurements, new laser beam positioning system, and plasma light monitor system for increasing data reliability. In this paper, we describe the recent progress of the LHD Thomson scattering system in detail.
Your cart is empty.
Home|Invoice Payment|Nuclear Links